Product Detail: |

# Multipolar Magnetic Field Distribution Tester

- DX-2100R Multipolar Magnetic Field Distribution Tester
- DX-2100RB Multipolar Magnetic Field Distribution Tester
- DX-2100RC Multipolar Magnetic Field Distribution Tester

Product Detail: |

# DX-2100RC Multipolar Magnetic Field Distribution Tester

### Introduction

DX-2100RC multipolar magnetic field distribution tester is developed and designed by Xiamen Dexing Magnet Tech. Co., Limited, combining with the actual needs of magnet ring manufacturers and motor companies, after years of unremitting efforts. It is high-tech product, specially used for the measurement and analysis of multipolar magnet ring surface magnetic field distribution. For magnet ring manufacturers and related motor enterprises, it is an indispensable tool for designing magnet ring and quality inspection.

**System configuration:**

**1. **Test platform: including electronic control regulating mechanism of probe, electronic control regulating system of XYZ, optic power-drive rotating platform and precise three jaw chuck.

**2.** Multipolar magnetic ring test host: including automatic range gauss meter and stepper motor drive control center.

**3. **DX-2100 Multipolar magnetic field distribution test software.

**4.** Lenovo computer and 1 set HP laser printer.

**5. **One set testing probe (easily-worn parts: sending 1 set).

**6. **Standard magnetic field: offer the inspection report of the third party (1000Gs).

### Technical index:

**1. **The displacement of probe X, Y, Z could be power-derived. The strokes of X, Y and Z are 300mm, 200mm and 200mm.

**2.** X, Y, and Z: the moving resolution is 0.01mm. The accuracy of return is ±0.02mm.

**3.** Rotating concentricity of three jaw chuck: 0.01mm. The whole concentricity is better than 0.05mm.

1). Optic power-drive translation platform, model: 01RS004 optic power-drive rotating platform, ratio of transformation: 1:180.

2). Three jaw chuck, model: SE-03 , precision measurement 36000 step/revolution, test time: 60s.

**4.** Test host, gaussmeter has 4 automatic ranges: 0.2K/0.4K/1K/2K/4K/10K/20KGs 7 automatic ranges, the resolution: 0.2%of full range for each type range, below 400Gs, 0.1Gs is effective, accuracy: 1.0%of full range for each type rang.

**5.** Test probe: 0.65mm thickness, 3.5mm width, the extended length of leading end is 60mm.

**6. **Soft ware:

1). Automatically measure maximum extreme value, minimum extreme value and average extreme value of each pole magnetic field.

2). Automatically measure maximum extreme value, minimum extreme value and average extreme value of each magnetic pole.

3). Automatic measure area of each magnetic pole and half-width.

4). Functions of zero self-return and getting rid of invalid data with anti-shake.

5). Automatically output XY graph curve and it could magnify at will on XY coordinate.

6). Automatically output polar diagram.

7). Automatically output corresponding fundamental wave and harmonic analysis.

8). Regarding the same sample, displaying three dimension magnetic field distribution by combining test points.

9). Storage function of lists of pole angle, pole extreme value and pole area.